Alcohol Meter [ALM-155]
Heartwarming story of father-daughter winery
A father sticking to traditional method, a daughter
introducing modern technology.
How can they go along with?
Dedicated website for ALM-155 Alcohol Meter
This is special website only for introduction of ALM-155 Alcohol Meter. Please visit the website.
Kyoto Electronics Manufacturing Co., Ltd., has valuable experience and long history in the manufacturing of oscillation density technology from 1978. The technology of ALM-155 origins from our Sake alcohol analyzer DA-155 which has been sold mainly to Japanese Sake breweries for many years. Most of the Sake breweries are small family businesses, however, their demand for reliable analytical instrumentation is very strong. Therefore, KEM has been providing them with easy to use, high performance analyzer at a reasonable price.
The ALM-155 is a dedicated, small size & high-performance bench top density meter with oscillating capillary tube for the analysis of oenology, sprits and beer. The analysis of Alcohol, density and specific gravity in alcohol products has never been so easy and accessible for any budget. The ALM-155 displays results with high-resolution of 0.01% for alcohol content and 0.00001 g/cm3 for density. Moreover, with the peltier thermostat system, the temperature is fixed at exactly 20°C/60°F.
One button measurement with small sample volume
The ALM-155 is equipped with peristaltic pump for automatic sampling. The required sample volume is only 8mL and can be used for repeatability testing with the automatic repeating function. The analysis of alcohol content (on the distilled wine sample) or density (on wine as it is) is performed in two to four minutes for one measurement by pressing just one button. When high viscous sample is measured, an optional syringe can be utilized to introduce the sample into the measurement cell what makes the ALM-155 suitable for the analysis of any alcohol products.
Simple calibration with pure water and easy maintenance
The calibration of ALM-155 does not require expensive standards, only pure water. After analysis, the unit can be cleaned by simply washing the cell with pure water or a kitchen detergent. The only replacement parts are the sampling tubes and those can be replaced by the user. The daily performance check can be easily performed by the operators and requires pure water only.
Automatic Saving results with digital data
The results are not necessary to be recorded by hand-writing. An optional printer can automatically print out the results once the measurement is completed. The ALM-155 can also store measurement up to 100. The data can be transferred to a connected USB memory stick or to an optional PC software through an RS-232C cable.
Alcohol is the main legal parameter for bottled wines; the declaration of fake values entails severe penalties for wine producers. For this reason, the analysis of wine must be done in accordance with the official analysis method.
It is very important to check density, specific gravity and alcohol content during wine fermentation, in order to verify its progress and provide notice if it stops prematurely.
The density of grape juice indicates the amount of sugar contained in the juice before fermentation. It can determine the expected alcohol content in wines, so it is very important to analyze its density accurately.
In addition to providing analytical support to wineries, these laboratories can release certificates of analysis with legal validity. As well as wineries, these laboratories have to perform analyses in accordance with the official method in order to be sure of the accuracy of the values they certify.
Distilleries and spirits…
Companies producing spirits, such as distilleries, have products with a high alcohol content. The ALM-155 is a high performance instrument working in the whole concentration range from 0.00% to 100% alcohol content. Therefore, the instrument is highly suitable for this application.
Monitoring density and specific gravity during the beer production process is very important for the quality of the final product. ALM-155 can provide alcohol testing after the beer sample has been distilled.
Vinegar is obtained by oxidation of the ethanol content of wine, cider, beer, other fermented alcoholic beverages, or raw materials such as malt, rice and raw and cooked fruit. Checking the alcohol content during the oxidation process is very important to understand when ethanol has completely turned into acetic acid.
|Model||ALM-155 Alcohol Meter|
|Measurement Range||Alcohol content||0.00～100.00 vol%|
|Repeatability||Alcohol content||SD：0.05 vol%|
|Resolution||Alcohol content||0.01 vol%|
|Measurement Temperature||20°C / 60°F|
|Alcohol Table||OIML（vol%/wt%）/ HMCE（At 20℃） / G225（At 20℃） / AOAC / NIST(At 60°F)|
|Calibration||Calibration：One point with pure water or two points with dry air and pure water (Note: dry air pump should be prepared locally for two points calibration)|
|Min. Sample Required Automatic sucking in||ca. 8mL (for 10 seconds ofsampling time)|
|Measurement Time In auto operation||2 to 4 min. (with the peristaltic pump)|
|Display||Display Instrument||LCD (with a backlight)|
|Displayed contents||Item||5-2 mode||4-1 mode|
|Sampling||1) Auto Sampling by Peristaltic Pump
2) Manual by optional syringe
|Auto Start function||Number of repeat time: 2-100|
|Number of Measurement Result||100|
|Density-Concentration Conversion||Alcohol content
Fixed（Stored as Standard）
|PC Software||SOFT-CAP (Data Acquisition Software)|
|Auto Cleaning Unit||Not Connectable|
|Multiple Sample Changer||Not Connectable|
|Interfaces||RS-232C||Dot Matrix Printer (COM1) , PC (COM2)|
|USB||USB flash drive , keyboard|
|Data I/O||CSV data saving to USB|
|Ambient Conditions||1) Temperature ： 5 to 35ºCM
2) Humidity ： Below 85%RH（no condensation）
|Power Supply||100 to 240 VAC±10%, 50/60Hz|
|Power Consumption||Approx. 30W|
|Dimension||270 mm (W)×402 mm (D)×163 mm (H)|
|Weight||Approx. 10 kg|
|onformity standard||CE marking EMC: EN61326-1 LVD: EN61010-1